Timepix, Computed Tomography, Dose
||August 07, 2017
||The Author(s) 2017. This article is published with open access at www.chitkara.edu. in/publications
Research using hybrid pixel detectors in medical physics is on the rise. Timepix detectors have arrays of 256 × 256 pixels with a resolution of 55 μm. Here, and by using Timepix counts instead of Hounsfield units, we present a calibration curve of a Timepix detector analog to those used for CT calibration. Experimentation consisted of the characterization of electron density in 10 different kinds of tissue equivalent samples from a CIRS 062M phantom (lung, 3 kinds of bones, fat, breast, muscle, water and air). Radiation of the detector was performed using an orthodontic X-ray machine at 70 KeV and .06 second of tube current with a purpose-built aluminum collimator. Data acquisition was performed at 1 frame per second and taking 3 frames per phantom. We were able to find a curve whose behavior was similar to others already published. This will lead to the verification of the usage of Timepix for identification of different tissues in an organ.
- Franca Cassol Brunner: First K-Edge Imaging With a Micro-CT Based on the XPAD3 Hybrid Pixel Detector. IEEE Transactions on Nuclear Science 60(1), 103–108 (February 2013)
- P. Delpierrea, (February 2007) e.: PIXSCAN, Pixel detector CT-scanner for small animal imaging. Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment(1-2), 425–428.
- CERN: Medipix. Available at: https://medipix.web.cern.ch/medipix/pages/medipix2/timepix.php
- M. Esposito, e.: Energy sensitive Timepix silicon detector for electron imaging. Nuclear Instruments and Methods in Physics Research A (October 2011)
- J. Seco, (February 2006) P.: Assessing the effect of electron density in photon dose calculations. MEDICAL PHYSICS 33(2).
- J. Jakubek: (August 2009) Energy-sensitive X-ray radiography and charge sharing effect in pixelated detector. Nuclear Instruments and Methods in Physics Research 607(1).
- Zemlicka: Energy and position sensitive Pixel detector Timepix for X-ray fluorescence imaging. Nuclear Instruments and Methods in Physics Research 607(202) (2009)
- J. Jakubek: Pixel detectors for imaging with heavy charged particles. Nuclear Instruments and Methods in Physics Research 591(1) (2008)
- J. Jakubek: A coated pixel device TimePix with micron spatial resolution for UCN detection. Nuclear Instruments and Methods in Physics Research 600 (2009)
- Boog, R.: Energy calibration procedure of a pixel detector. (2013)
- Campbell, M.: Charged particle detection using the timepix and timepix3 chips and future plans. (Accessed 2012) Available at: https://www2.physics.ox.ac.uk/sites/default/files/2012-03-27/mcampbell_ oxford_pdf_14057.pdf
- GNATUS Available at: http://www.gnatus.com.br/site/esp/produtos_show. php?id=1133&cat=813&scat=imagen
- CIRS Available at: http://www.cirsinc.com/products/all/24/electron-densityphantom/
- Advacam In: Advacam. Available at: http://www.advacam.com/en/products/ fitpixkit
- A. Butler, P.: Measurement of the energy resolution and calibration of hybrid pixel detectors with GaAs:Cr sensor and Timepix readout chip. Physics of Particles and Nuclei Letters 12(1) (January 2015)