Development and Validation of an X-ray Imaging Detector for Digital Radiography at Low Resolution
Digital X-ray detectors are required in different sciences and applications, however many high quality devices are expensive although high-resolution images are not always required. We present an easy way to build a detector capable of forming X-ray digital images and video with a very large area (18×18 cm2). The detector is formed by three main components: scintillator, optics lenses and CCD sensor. Basically, the device converts the X-rays into visible light which is then collected by the CCD sensor. The scintillator is Gadox type, from Carestream®, 18×18 cm2, regular type, lambda 547 nm. The optics lenses are generic, with manual focus and widely visual field. The CCD sensor has a size of 1/3″, 752 × 582 pixels, monochrome, 20 FPS, 12 bits ADC and pixel size of 3.8 μm. With the built detector and an X-ray source, we formed an X-ray imaging detection system to generate digital radiographs of biological or inert objects-examples are given-, as well as real-time X-ray video. Additionally, the spatial resolution limit was measured in terms of Modulation Transfer Function by the method of opaque edge from a lead sheet with a result of 1.1 Lp/mm. Finally using a filter, the focal spot of the X-ray source is measured, resulting in a diameter of 0.9 mm (FWHM).
A. B. Reed, Journal of Vascular Surgery 53, 3S (2011). https://doi.org/10.1016/j.jvs.2010.07.024
H.E. Martz et al., X-ray Imaging: fundamentals, industrial techniques and applications. CRC Press, (2016).
T. Angsuwatanakul et al., International Symposium on Communications and Information Technologies, Bangkok, 2006, 1145 (2006). https://doi.org/10.1109/ISCIT.2006.339959
M. Nikl, Meas. Sci. Technol. 17, R37 (2006). https://doi.org/10.1088/0957-0233/17/4/R01
A. D. A. Maidment and M. J. Yaffe, Phys. Med. Biol. 40, 877 (1995). https://doi.org/10.1088/0031-9155/40/5/011
A. D. A. Maidment and M. J. Yaffe, Phys. Med. Biol. 41, 475 (1996). https://doi.org/10.1088/0031-9155/41/3/010
Zschornack, Günter H. Handbook of X-ray Data. Springer Science & Business Media, (2007).
Knoll, Glenn F. Radiation detection and measurement. John Wiley & Sons, (2010).
Als-Nielsen, Jens, and Des McMorrow. Elements of modern X-ray physics. John Wiley & Sons, (2011). https://doi.org/10.1002/9781119998365
J. G. Rocha and S. Lanceros-Mendez, Recent Patents on Electrical & Electronic Engineering 4, 16 (2011). https://doi.org/10.2174/1874476111104010016
E. Samei, M. J. Flynn and D. A. Reimann, Medical physics 25, 102 (1998). https://doi.org/10.1118/1.598165
Hartmut Illers et al., Radiation Protection Dosimetry 114, 214 (2005). https://doi.org/10.1093/rpd/nch506
J. D. Everson and J. E. Gray, Radiology 165, 261 (1987). https://doi.org/10.1148/radiology.165.1.3628780
B. A. Arnold, B. E. Bjarngard and J. C. Klopping, Physics in Medicine & Biology 18, 540 (1973). https://doi.org/10.1088/0031-9155/18/4/006
B. K. Cha et al., Radiation Measurements 45, 742 (2010). https://doi.org/10.1016/j.radmeas.2009.12.025
Copyright (c) 2020 Abdiel Ramírez Reyes et al.
This work is licensed under a Creative Commons Attribution 4.0 International License.
Articles in Journal of Nuclear Physics, Material Sciences, Radiation and Applications (J. Nucl. Phy. Mat. Sci. Rad. A.) by Chitkara University Publications are Open Access articles that are published with licensed under a Creative Commons Attribution- CC-BY 4.0 International License. Based on a work at http://jnp.chitkara.edu.in. This license permits one to use, remix, tweak and reproduction in any medium, even commercially provided one give credit for the original creation.
View Legal Code of the above-mentioned license, https://creativecommons.org/licenses/by/4.0/legalcode
View Licence Deed here https://creativecommons.org/licenses/by/4.0/
|Journal of Nuclear Physics, Material Sciences, Radiation and Applications by Chitkara University Publications is licensed under a Creative Commons Attribution 4.0 International License.
Based on a work at https://jnp.chitkara.edu.in/