Implementation of a Low Cost For Internal Dosimetry Radiological Survey of Ionization Radiation Exposed Workers In Nuclear Medicine Services
Because the nature of the routine activities carried on in nuclear medicine services, where several not sealed radioactive sources are used, the occupational personal exposed to ionization radiation (POE) is exposed to both, internal and external radioactive contamination and direct radiation also. The most important radioisotope from the point of view of radiological risk in such Nuclear Medicine Services (NMS) is the 131I. The Mexican official regulatory organization in radiological and nuclear safety: Comisión de Seguridad Nuclear y Salvaguardias (CNSNS) specify, as a requirement in the operation license of Nuclear Medicine Services the mandatory of internal and external dosimetry survey of POE. The external dosimetry survey is normally carried out but the internal dosimetry survey is not easy to do, because internal dosimetry systems are too expensive and in Mexico there are only three organizations (CFE-CLV, ININ, CNSNS) where there are these systems but they aren ́t easily available for the Nuclear Medicine Services. Because that before fact, we propose a single, and low cost system (SLCS) to survey the internal dosimetry of POE. This system is based in a scintillation NaI(Tl) 2x2 detector and a common radiation monitor tipically used in the routine radiation and contamination measures, with a minimal infrastructure investment. This system is numerical calibrated with Montecarlo method, and compared with an well established commercial System being a good function corresponding. This system is easy to implementing in any nuclear medicine to complain the requirement of operation license in the internal dosimetry POE survey.
Berger, M.J, (1963). Monte Carlo Calculation of Penetration and Difussion of fast charged particles.
Bogi B,(2015). International Dosimetry Exercise for Enhanced Ability
Bronson F.Z, (1996). Validation of the MCNP Monte Carlo code for Germanium detector gamma efficiency calibration. Tuccon.
Eberline, (1991). Smart Radiation Technical Manual model SMR-200
IAEA, (1999). Assessment of Ocupational Exposure Due to Intakes of Radionuclides. Report No: RS-G1.2.
ICRP, (1977). Limits For Intakes Of Radionuclides Wokers. Report No:30
Knoll, GF, (2000). Radiation Detection and Measurement
Laboratory LAN (2008). MCNP user´s manual. Los Alamos National Laboratory.
Reglamento General de Seguridad Radiológica, (1988), (pp. 1-60). México: D.F.
Diago, J,R, (2005). Simulation of Detector calibration using MCNP.
Copyright (c) 2017 Journal of Nuclear Physics, Material Sciences, Radiation and Applications
This work is licensed under a Creative Commons Attribution 4.0 International License.
View Legal Code of the above-mentioned license, https://creativecommons.org/licenses/by/4.0/legalcode
View Licence Deed here https://creativecommons.org/licenses/by/4.0/
|Journal of Nuclear Physics, Material Sciences, Radiation and Applications by Chitkara University Publications is licensed under a Creative Commons Attribution 4.0 International License.
Based on a work at https://jnp.chitkara.edu.in/